Task Group 142 (TG-142) of the American Association of Physicists in Medicine (AAPM) recommends that the collimator, couch, and the gantry rotation isocenters should be checked against baseline with ±1 mm tolerance during annual Quality Assurance (QA) in the Mechanical subsection of Table III. The Linac isocenter diameter is evaluated with "star-shot" pattern formed by a radio-opaque object on the couch that is exposed to an open radiation field, measured on an Electronic Portal Imaging Device (EPID), as the table is rotated around the isocenter. Below is an example set of a star shot images:
The software can combine the images into a single image:
Ideally the lines would all intersect in a single point but in practice the lines intersect geometrically in a small volume that looks more like a sphere or ellipsoid. The center of this sphere or ellipsoid is the radiation isocenter. Below is an example plot:
The radiation isocenter may not coincide with the mechanical isocenter. Different beam energies may yield different isocenters.
The Total QA analysis software performs the following measurements:
- Minimum Tangent Circle Plot
- Table Rotation Isocenter
The measurements are completely automated, requiring the user only to drag and drop the file into the web-based software interface. A detailed report is created.
Table Starshot files must be DICOM files and contain the "starshot" within the file name. The minimum line width, as formed by the radio-opaque object, must be at least 4mm as determined by the resolution in the file.
Recommended Total QA tests are Annual Linac QA [TG-142 Table III]: 'Table rotation isocenter (starshot test)' and 'Minimum tangent circle plot (table starshot)'.
When imaging QA tests are added to templates an upload control will appear in the scheduled QA's data entry screen allowing the user to upload images for automated analysis.
To add files to the upload queue simply drag them from a Windows Explorer folder to the drag and drop folder and release them. Alternatively, by clicking on the Add Files button to the lower right of the control a windows file selection dialog will open and files can be selected for upload. Under either method, multiple files may be selected for upload at once.
If the automatically upload checkbox is checked(the default) then file uploading will start immediately as files are added
If the automatically upload button is turned to off the file upload process must be started manually clicking the Start upload button on the lower right of the control. To clear the upload queue click the Clear button.
Once file series have been uploaded they will be displayed below the upload control.
To remove a series from the queue click the Cancel button beside the series. To start processing click the Start Processing button. A description for the image series can be added at this point. Click the Edit button next to the series. Type a description for the series into the text box that appears below Description and either click Save or press the enter key. The description can also be edited after the images have been processed. Descriptions will appear in the report with the analysis of the series.
While files are being processed users may perform other tasks such as data entry.
|Annual Linac QA [TG-142 Table III]||Mechanical||Table rotation isocenter (starshot test)|
|Annual Linac QA [TG-142 Table III]||Mechanical||Minimum tangent circle plot (table starshot)|
Specifications to Create a Plan File
When creating a plan file to generate the Table Starshot, the beam width, must be a minimum of 4mm as determined by the resolution in the file and each beam must be distinct.
Detailed Discussion of Measurements
Below is an example of the report generated from the starshot tests:
Below is an example table star shot file set for use in testing: