Topics covered in this article include:
- Measurement Summary
- Available Tests
- Specifications to Create a Plan File
- Example Plan File Downloads
- Detailed Discussion of Measurements
- Example Files
Task Group 142 (TG-142) of the American Association of Physicists in Medicine (AAPM) recommends verifying MLC transmission (average of leaf and interleaf transmission) annually in Table V. Multileaf collimation.
The static leaf transmissions are calculated from comparisons between open and closed fields in both the A and B leaf banks. For the dosimetric leaf gap calculation a series of increasing leaf gaps is swept across a 10x10 cm field. The EPID readings for each gap are then corrected and regressed to calculate the leaf gap.
Below is an example test image:
The analysis displays the following test results:
- A and B Leaf Bank MLC Transmission Percentage
- Dosimetric leaf gap
The measurements are completely automated, requiring the user only to drag and drop the file into the web-based software interface. A detailed report is created.
The images must be DICOM files and must contain "mlctrans" somewhere in the file name. Alternatively, limited capabilities to manually identify planar images using DICOM tag values have been added to the image processing system. This is an extension of the existing naming convention system. The PatientID, StudyID, and SeriesDescription DICOM tags are checked and if the text "mlctrans" (case insensitive) is found the image set will be processed as an asymmetric field test series. For more details see Manual Identification of RT Planar Images and Individual Catphan Slices through DICOM tags.
When imaging QA tests are added to templates an upload control will appear in the scheduled QA's data entry screen allowing the user to upload images for automated analysis.
To add files to the upload queue, simply drag them from a Windows Explorer folder to the drag and drop folder and release them. Alternatively, by clicking on the Add Files button to the lower right of the control a windows file selection dialog will open and files can be selected for upload. Under either method, multiple files may be selected for upload at once.
If the automatically upload checkbox is checked (the default) then file uploading will start immediately as files are added.
If the automatically upload button is turned to off the file upload process must be started manually by clicking the Start upload button on the lower right of the control. To clear the upload queue click the Clear button.
Once file series have been uploaded they will be displayed below the upload control. To remove a series from the queue click the Cancel button beside the series. To start processing click the Start Processing button. A description for the image series can be added at this point. Click the Edit button next to the series. Type a description for the series into the text box that appears below Description and either click Save or press the enter key. The description can also be edited after the images have been processed. Descriptions will appear in the report with the analysis of the series.
While files are being processed users may perform other tasks such as data entry.
Annual MLC QA [TG-142 Table V]
Multileaf collimation [TG-142 Table V]
Dosimetric leaf gap
Dosimetric leaf gap readings
Dosimetric leaf gap plot
Images of open field and transmission banks A and B
Specifications to Create a Plan File
Please use your manufacturers recommended plan file.
Detailed Discussion of Measurements
Below is an example of the test output generated on a report:
Below are example DICOM files for use in testing: